AGILENT E3640A LABVIEW DRIVER
Message 11 of Message 12 of Thanks and sorry again. To be sure I had things working correctly, I first connected the power supply to an “on board” comport and used Tera Term to verify I had the correct cable, baud rate and such. If it is working and you take the computer end of your cable and transfer it to RS end of your USB comport, change the com port ref and it fail to send and receive the same commands, Then the USB to RS are not compatible with your EA. Not without problems, but generally they work.
|Date Added:||9 June 2015|
|File Size:||29.41 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Agilent | Agilent E36xx IVI-COM Driver
You can download the parser and its associated SDK from: Do you have the correct combination of NULL modem, gender changer and cable? If not found, the installer will abort, warn, or install the required component as appropriate. Is it normal that remote control is that much slower than manual control? Message 17 of They may agilenh be “broken”, but may not be designed to probvide the correct signal voltage levels for your power supply.
Sorry, I am quite new to the world of online communities, and have made a blunder by crossposting. That made the initialize sub VI run. The supported model numbers are: You can also install this driver from another installation package.
Agilent Technologies controllers generally agielnt address. Accepted by topic author Keksbold. After I switched this setting on the device, I was able to set it via RS These three strings are values of properties in the IIviIdentity interface.
Message 18 of What kind of USB – serial device are you using? More Information For more information about this driver and other instrument drivers and software available from Agilent Technologies visit: They both don’t work and as they are quite new, I am pretty sure they are not broken.
Agilent E36xx IVI-COM Driver
It is able labviw install many of the required system components, including Windows Installer 2. Spoiler Highlight to read You are not the first person to forget the front panel setting for external control source! I somehow think it would be a bad idea to just disable all the stuffs and hope fopr the best Install The installation package for the driver is distributed as an MSI 2.
Just as a sanity check. Thank you very much for your help, Keksbold.
I fear this might not be fast enough for my system. The remote interface can be selected from the front-panel only. If you see an error about wscript.
Solved: Agilent EA – Page 2 – Discussion Forums – National Instruments
So my first fault were the front panel settings, and my second one was a missing NULL modem. Message 19 of Thank you very much for the example for setting the default values myself. Try another USB to Serial.
Not without problems, but generally they work. You can download the latest version from: If you plan to use the driver with. NET Framework itself is not required by this driver.